- July 22, 2019
-- Guide to Implementing Automated Temperature Monitoring
Personnel installing new temperature monitoring systems (https://www.dataloggerinc.com/
monitoring/) in their refrigerators and freezers are often stumped as to why the temperature appears to vary unexpectedly. Our data loggers provide significantly faster response time and significantly more data than conventional thermometers which often leads to the observation of temperature behavior that was previously missed. We have put together this quick guide to recommend some solutions and help you investigate so that over time you can identify the source and control the rhythm of your fridge or freezer.1. Frequent Door Opening
The first and simplest cause of unstable temperature readings occurs whenever people open the door, which almost immediately causes a change in your refrigerator/
freezer's internal temperature. Electronic temperature monitors have much faster response times and will display random jumps in temperature as much as 5-10 degrees that may be interpreted as a problem with the system. If the presence of these jumps in temperature is an issue for your organization, you can add a thermal buffer (https://www.dataloggerinc.com/products/data-logger-access...
), which can be a small bottle of glycol or a nylon block, to the probe. This provides thermal mass which will dampen the system's response time and eliminate the temperature spikes caused by opening the door.2. Compressor Cycling
The second source of temperature variation arises from the normal cycling of the compressor used to cool the refrigerator/
freezer. By looking at a graph of temperature vs. time, this cycling will be revealed by a regular, periodic change in temperature. We occasionally find the probe for the monitoring system placed very close to the evaporator coil or directly in the airstream of the coil fan. These locations aren't ideal since they experience the greatest change in temperature, so relocating the probe to a spot away from the coil or out of the airstream will usually help reduce variation amplitude.3. Electrical Noise
The final cause is due to electrical noise in the monitoring system. This can often be identified from the temperature data in the form of random jumps in temperature. The most common cause of electrical noise is the placement of wires from the sensor. If the wires carrying these signals pass near to wires carrying line voltage power signals for motors, it's easy to get stray voltage induced by the temperature signal from the sensor. Ways to fix this are to carefully place the temperature monitoring sensor wires away from wires carrying AC power, use twisted pair wiring (https://en.wikipedia.org/wiki/Twisted_pair
), use shielded temperature monitoring sensor wires, and ensure the shielded cable is connected to a good earth ground free of potential ground loops.
Utilizing these suggestions with your fridge monitoring system, you should be able to minimize or remove the majority of temperature fluctuations in your refrigerator/
For more information, visit www.DataLoggerInc.com.