News By Tag
News By Place
Macro Sensors LVDT Linear Position Transmitters Gauge Tank Volume Levels
Macro Sensors LVDT Linear Position Transmitters Serve as Level Sensors In Gauging Tank Volume Level Changes
In this LVDT level sensor configuration, a stainless-steel float coupled to a nonmagnetic stainless-steel rod is attached to the high-permeability, armature core of the LVDT. The 4 to 20-mA loop-powered LVDT (http://www.macrosensors.com/
In operation, as water level changes, the float moves up or down, raising qflbg the LVDT core along with it. A threaded stainless-steel rod, that protrudes from the other end of the core, carries two jam nuts that are used to adjust the position sensor output at the desired low water level. The nuts also prevent the core from falling out of the LVDT housing if the liquid level drops too low. A nonmagnetic clamp block attaches the stainless-steel body of the LVDT to a flat hook. The hook hangs the level-sensor assembly over the tank edge.
Core position is sensed using magnetic induction. Electronics of the LVDT are hermetically sealed inside a stainless-steel housing for protection against water and other environmental elements. A Teflon liner inside the bore minimizes any friction to core motion. As there are no springs to fatigue or parts to wear out, the LVDT Linear Position Sensors are nearly friction free to offer infinite resolution and long term operation.
For more information on this special configuration, contact the factory at positionsensors@